|Reading Time||5 minutes||Shelf Life||3 Years|
Individually packed test device,
Package insert, Dropper, Buffer
|Accuracy||＞99.9%||Feature 1||High Accuracy|
|Feature 2||High Sensitivity & Accuracy||Feature 3||Clear Results|
|Feature 4||Easy To Read||Feature 5||Easy To Use|
|Feature 6||OEM/ODM Available||Feature 7||Quality Guarantee|
Bring tests, specimens, buffer, and/or controls to room temperature (15-30°C) before use.
1.Bring the pouch to room temperature before opening it. Remove the test cassette from the sealed pouch and use it as soon as possible.
2.Place the test cassette on a clean and level surface.
3.Holding the sample collection tube upright, carefully take off the tip of collection tube, then break off the tip of collection tube, transfer 3 drops (approximately 100μl) to the specimen well(S) of the test cassette, then start the timer. See illustration below.
4.Wait for the colored line(s) to appear. Read results at 5 minutes. Do not interpret the result after 10 minutes.