|Reading Time||10-15 minutes||Shelf Life||3 Years|
Individually packed test device,
|Accuracy||＞99.9%||Feature 1||High Accuracy|
|Feature 2||High Sensitivity||Feature 3||Clear Results|
|Feature 4||Easy To Read||Feature 5||Easy To Use|
|Feature 6||OEM/ODM Available||Feature 7||Quality Guarantee|
Instructions must be read entirely before taking the test. Allow the test device controls to equilibrate to room temperature for 30 minutes (20°C -30°C) prior to testing.
Strip and Cassette:
1. Take off the outer packing, put the strip/cassette onto the desk with the sample adding area of the strip/ the sample window of the cassette up.
2. Serum/Plasma/Whole blood: Drop 1.5μL serum/plasma or 2μL whole blood vertically onto the membrane. The pipette tip of micro pipette or sample loop is required to touch the membrane gently for an accurate operation when the sample is added. If a disposable plastic sample ring is used, the ring body should be completely immersed in the sample, and then the membrane surface of the ring body vertical touch reagent plus sample should be sucked into the sample, and the membrane surface of the sample should have wetting marks.
3. Add about 2 drops of (80μL-100μL) sample dilution onto the sample pad of strip/the dilution well of cassette. Observe the test results immediately within 15-20 minutes, the result is invalid over 20 minutes.