|Reading Time||10-15 minutes||Shelf Life||3 Years|
Individually packed test device,
|Accuracy||＞99.9%||Feature 1||High Accuracy|
|Feature 2||High Sensitivity||Feature 3||Clear Results|
|Feature 4||Easy To Read||Feature 5||Easy To Use|
|Feature 6||OEM/ODM Available||Feature 7||Quality Guarantee|
Instructions must be read entirely before taking the test.
Allow the test device controls to equilibrate to room temperature for 30 minutes (20°C-30°C) prior to testing.
1.Remove the test cassette from the sealed pouch, place it on a clean and level surface with the sample adding area up.
2.Using the dropper, vertically transfer two drops (~50µL) of the whole blood into the sample well (S) of the cassette, avoiding the formation of bubbles. After the sample is absorbed, add 1 drop(40-50µL) of sample diluent into the sample well of the cassette. 3.Observe the test results immediately within 20 minutes, the result is invalid over 40 minutes.